What is the difference between Newheek’s NK103 and NK202 Collimator?
There are three kinds of medical tube collimator under the Newheek company. The following are the differences between the two NK103 and NK202 collimator devices.
First, NK103 is smaller than NK202 collimator in size, with a WLH of 158 mm x 138 mm x 83 mm and a WLH of 185 mm x 198 mm x 145 mm. In terms of weight, the weight of NK103 is 2.6kg, and the weight of NK202 is 6.8kg. In appearance, the NK103 dashboard knob is on different sides, and the NK202 dashboard knob is on the same side. NK103 is suitable for 125KV tube, while NK202 is suitable for 150KV tube. The maximum irradiation field of NK103 is 430 mm x 430 mm (SID = 90 cm), and that of NK202 is 440 mm x 440 mm (SID = 100 cm). The average brightness of NK103 is >100lux, and the average brightness of NK202 is >160lux. NK103 was >3:1 and NK202 was >4:1. NK103’s bulb 12V/50W, NK202’s bulb is 24V/150W.
Author:X Ray Collimator