How does the beam luster work? Let’s take a look today. X -ray beam lights are mainly connected to the ball pipe. The main role is to perform positioning during swinging. Simulation of the radiation area of X -rays can reduce the radiation dose of patients and enhance image quality. Its internal structure is also equipped with a lighting field indicator system. It uses the indicator light bulb to simulate the focus of the X -ray tube, instead of X -rays with visible light, and enter the bed surface after reflection is reflected by the reflex mirror. After reflection, the light path that can be visible is consistent with the light path after the reflex mirror, which can indicate the size of the irradiation in advance. Due to the short service life of the indicator light, the lighting of the indicator light is controlled by the button switch, which can be automatically closed within 15S to 1min. Some indicators have two levels of brightness in the service life to meet the requirements of the interior high brightness lighting site instructions. Rephue is usually made of silver -plated plastic. The amount of aluminum is about 0.5 mm, which will not affect X -ray imaging. A clean and transparent plastic board is also installed in front of the beam limiter, which is engraved with black lines or black spots on the plate, indicating the center of the light field. Some light beam limits are marked with the corresponding focal length of the irradiated wild size around the knob or the range of the tie rod activity. It is mainly used to set up the light wild size before lighting the light indicator light to light up, so that the indicator light is turned on quickly. Extend the life of the indicator light.This is the working principle of X -ray beam light . If you are interested, welcome to call! Tel: +8617616362242
Author:X Ray Collimator
Company:Weifang Newheek Electronic Tech Co., Ltd.
ADD:E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China