With the development of the X ray machine, the x ray collimator has evolved from simple to complex in structure, with functions ranging from single to multi-purpose. Manual proper collimation of the useful beam is mostly used in radiography. The proper collimation of the useful beam is achieved by manually adjusting the knob or the rod on the proper collimation of the useful beam. The internal structure of the proper collimation of the useful beam is provided with an illumination field indicating system in addition to the foregoing.
Introduce Newheek manualproper collimation of the useful beam 202 today.
It is designed for different x ray machine whose x ray tube is 150kV. And the max x ray imaging size is 17”x17”.
The suitable x ray tube max voltage: 150 kV
Max irradiation field: SID (FFD) = 1000mm hours, no less than 430 mm x 430 mm
Min irradiation field: 0x0 mm
Inherent filter: 1 mm·Al/75kVp
Additional filter: 1 mm·Al and 2 mm·Al
Lead leaf control mode: manual
Working power: 24V AC/DC, 150 W (The lowest safe voltage)
If you are interested in our Newheek x ray collimator, please contact our customer service.