What is the role of the DR beam limiter? The beam limiter uses a lead plate with an adjustable gap to block the unnecessary original emission lines emitted from the window, thereby controlling the size of the beam and changing the actual X-ray irradiation field. The relationship between the focus, the lead plate and the irradiation field A/B=a/b A is the distance from the focus to the lead plate; B is the distance from the focus to the imaging surface; a represents the opening and closing degree of the lead leaves; b represents the size of the irradiation field.
Introduce our beamer to customers
Uses: used for mobile X-ray machines, stationary X-ray machines, mainly matching 125 KV X-ray tubes
X-ray tube maximum voltage: 125 KV
Maximum irradiation field: 440 x 440 mm² (SID = 1000 mm)
Bulb: 24 V / 100 W
Light bulb single lighting time: 30 s
Lead leaf (optional): 1 layer
Lead Leaf Control Mode: Manual
Input power: 24 V AC
SID measuring tape with soft ruler: yes
Dimensions: 223 (W) x 185 (L) x 87 (H) mm
Weight (excluding cables): 5.5 kg
If you are interested in our beamer, please contact our customer service.
Author:X Ray Collimator