X ray collimator is a multi-purpose hand perspective device, which is convenient for long-distance control. It is an indispensable device for remote X-ray machine. The structure of the electric beam limiter is basically the same as that of the manual East limiter, except that the moving power of the shield is provided by a small motor. The irradiation field can be adjusted to the required size by controlling the motor’s reverse rotation and operation time. In general, micro DC motor is used in the motor, and a limit switch is set at the closing and maximum opening position to automatically limit the position. The movement is sensitive and easy to control. The electric beam limiter for perspective needs to adjust the size of the irradiation field at any time without the indication of the irradiation field and the light indication. However, the electric beam limiter, which is also used for fluoroscopy of photography, should be equipped with illumination field display and light indication system.
The difference between the electric beam limiter and the automatic beam limiter is that manual calibration is needed. The automatic beam limiter is a beam limiter that automatically regulates the closure of the lead leaf.
Company:Weifang Newheek Electronic Tech Co., Ltd.
ADD:E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China