Usually, the x ray collimator is available in both manual and electric. Manual for photography, electric for perspective. The principle structure of the two smalls x ray machines is basically the same, but the power of the adjustment is different. Today, I will mainly tell you about the manual limiter.
Manual small x ray machine directly manually open and close the hidden line of the small x ray machine. There are two types of operation: knob type and pull rod type. The adjustment knob or the adjustment lever is provided with a corresponding position illumination field size indicator scale. In this way, the required field size at a predetermined distance can be roughly adjusted before the illumination field indicator lights up. Correct use can shorten the lighting time.
The three small x xray machines that Newheek sells are manual. Let me introduce you to Newheek smallest x ray machine202:
It is designed for different x ray machine whose x ray tube is 150kV. And the max x ray imaging size is 17”x17”.
The suitable x ray tube max voltage: 150 kV
Max irradiation field: SID (FFD) = 1000mm hours,
No less than 430 mm x 430 mm
Min irradiation field: 0x0 mm
Inherent filter: 1 mm·Al/75kVp
Additional filter: 1 mm·Al and 2 mm·Al
Lead leaf control mode: manual
Working power: 24V AC/DC, 150 W (The lowest safe voltage)
If you are looking for smalls x ray machine collimator, please contact our customer service.
Company:Weifang Newheek Electronic Tech Co., Ltd.
ADD:E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China