With the development of the X ray machine, the x ray collimator has evolved from simple to complex in structure, with functions ranging from single to multi-purpose.
According to the driving method of the beam limiter opening and closing of the beam limiter, it can be divided into manual x ray collimator, electric x ray collimator and fully automatic x ray collimator.
Tell us about the electric X ray collimator today. The electric collimator is mostly used in fluoroscopy, which is convenient for remote control and is an indispensable component for the remote control of the gastrointestinal bed.
The opening and closing of the electric collimator lead leaf is generally driven by a micro DC motor. The forward and reverse time of the DC motor can be properly controlled, and the irradiation field can be adjusted to the required size. The adjustment of the electric collimator illumination field is carried out both on the collimator and on the console at the bedside. In addition to continuous adjustment, the latter has various fixed-size illumination field selection buttons, just press With these buttons, the motor is driven to move the lead leaf to the selected field and fixed to meet the specific requirements of the radiography. The motor operates with a limit switch at the lead leaf closing and maximum opening position, and automatic limit protection.
If you want to know more about the Newheek electric collimator, please contact our customer service.
Company:Weifang Newheek Electronic Tech Co., Ltd.
ADD:E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China