The first thing to understand is what X-rays mean? An X-ray generating device is a device capable of generating X-rays for clinical diagnosis. The X-ray generating device is mainly composed of a high voltage generator, a bulb and a Collimator. The X Ray Collimator is usually disposed under the bulb, and the X-rays emitted by the bulb are irradiated onto the object to be inspected by the beam limiter. The role of the Collimator is to limit the range of X-rays emitted from the tube. The Collimator generally includes a housing, a positioning lamp, a reflecting lens and an opening limiting structure, and the positioning lamp, the reflecting lens and the opening limiting structure are installed in the housing.
Tell us about Newheek x ray collimator NK202 today.
It is designed for different x ray machine whose x ray tube is 150kV. And the max x-ray imaging size is 17”x17”.
The suitable x ray tube max voltage: 150 kV
Max irradiation field: SID (FFD) = 1000mm hours,
No less than 430 mm x 430 mm
Min irradiation field: 0x0 mm
Inherent filter: 1 mm·Al/75kVp
Additional filter: 1 mm·Al and 2 mm·Al
Lead leaf control mode: manual
Working power: 24V AC/DC, 150 W (The lowest safe voltage)
If you want to know more about Newheek x-ray collimator, please contact our customer service.