In the past, we have mostly explained the use of X-ray beam optics, the role of X-ray collimator optics, and the scenes and connections of X-ray collimator optics. In this article, we will focus on explaining some techniques of X-ray collimator optics. Parameter problems, manual rotary button scale:
Focal screen distance: the distance from the focus to the screen or cassette
Focal skin distance: the distance from the focal point to the skin
1.1 meters: the distance between the focus of the tube and the photosensitive plane of the cassette (for gastrointestinal use)
1.5 meters: minimum chest radiography distance
1.8 meters: the tube is loosened (generally 45°, 35°, less than 60°), and the tube is played outward (such as hitting the chest frame)
For chest radiographs, the minimum is 1.5 meters, which is subject to 1.8 meters.
1.1m——70cm remote control, gastrointestinal, bed movement
Centimeters and inches are marked, corresponding to the size of the film, the effective range of rays.
After speaking so much, you have a deeper understanding of the product knowledge of the collimator beamer. Today’s sharing comes here first. In addition, our company specializes in the production of image intensifier TV systems. The products include image intensifiers, CCD cameras, and image processing. Devices, monitors, and products are exported to Europe, America, Africa, and the Middle East, and are well received by customers from various countries. In addition, our company specializes in producing all kinds of vertical and mobile chest frames, various types of x-ray machine parts,welcome to call Negotiations.