Does the optical device of the digital x-ray imaging system fail? Do you know how to repair this problem?
The beamer is an essential component of the digital X-ray system, and it is inevitable that some faults will be encountered during the use.
Let me share with you the previous repair cases. When the DR is turned on, the system is normal and cannot be exposed.
The fault analysis should be that the beamer of the digital x-ray imaging system is faulty. There is an abnormal clicking sound in the beamer, and it feels that there are parts inside the Dongguang device that are stuck. In order to see the problem inside the light beamer, we disassemble the shell of the light beamer. It is seen that the filter plate inside the beamer is faulty and cannot be lifted normally. After careful inspection, it was found that the copper sheets on the corners of the first filter plate were slightly curled, and hooked the second filter plate during the rising process. Troubleshooting: We use needle-nose pliers to gently straighten the bent and deformed corners of the filter plate to restore it to its original shape, close the beamer cover, and the machine exposes normally. Discussion: Since the filter plate inside the Dongguang device moves up and down in turn, the chamfering of the first copper sheet drives the second copper sheet during lifting, and so on. If the driving force is slightly deviated, the chamfering of the copper sheet will cause | deformed. If the operator finds a slight abnormal noise of the motor in daily use, he can check and adjust the working environment of the equipment to prevent the further development of copper sheet deformation and aggravation, resulting in abnormal operation of the beamer. About the failure of the beam optical device of the digital x-ray imaging system, I will share with you today. If you need further cooperation, please contact us.
Author:X Ray Collimator