After purchasing the x ray collimator, we need to debug the x ray collimator for 125kv x ray machine in order to use it normally. What problems should we pay attention to when debugging? Next, let’s get to know!
Firstly, we understand the adjustment of the light bulb in the x ray collimator for 125kv x ray machine: the longitudinal position, that is, the distance from the reflector, the distance is far, the light field is smaller than the ray field, and the distance from the near field is larger than the ray field. If the transverse position is not on the central line of the reflector, the consistency of the light field and the ray field is deviated in the transverse direction. The angle of the bulb itself, here refers to the angle of the bulb filament relative to the reflector, which affects the size of the focus of the filament projected to the reflector, and the clarity of the cross line. Similar to the size of the focus projected by the bulb filament on the target surface, the smaller the focus, the clearer the cross line, the larger the focus, and the more blurred the cross line. Reflector debugging of x ray collimator for 125kv x ray machine: Angle affects field consistency. Reflectivity affects brightness.
Company:Weifang Newheek Electronic Tech Co., Ltd.
ADD:E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China