Let us first understand what components in the X-ray collimator are? Look at the specific debugging method!
Commissioning of the bulb: the longitudinal position, that is, the distance from the mirror, the distance is far, the light field is smaller than the radiation field, and the near light field is larger than the radiation field. The lateral position, if not on the center line of the mirror, the uniformity of the light field and the radiation field is deviated in the lateral direction. The angle of the bulb itself, here refers to the angle of the filament of the bulb relative to the mirror, affecting the size of the focus of the filament projected onto the mirror, affecting the sharpness of the crosshair. Similar to the case where the bulb filament is projected to the focus of the target surface, the smaller the focus, the clearer the crosshair, the larger the focus, and the more blurred the crosshair. Mirror adjustment of the X-ray collimator: The angle affects the uniformity of the light field. Reflectivity affects lightness.
If you are interested in our X-ray collimator, please contact us.
Company:Weifang Newheek Electronic Tech Co., Ltd.
ADD:E Building of Future Star Scientific Innovation Industrial Zone of No.957 Wolong East Street, Yulong Community, Xincheng Sub-District Office, Weifang Hi-tech Zone, Shandong Province, China